- Easy selection of prosthetic components in the laboratory
- Optional shortening of the impression post is possible
- Trouble-free impression of heavily disparallel implants
Open tray impression posts for making an impression at the implant level. The open tray impression method enables the most accurate impression to be made.
The fastening pin can be shortened by 3.5 mm, if necessary. There is a ring marked on the pin where it can be broken with the help of a screwdriver. The body of the impression post can be shortened with a separation disc.
Warning! The body of the impression post and the pin can only be shortened extraorally.
The pin can be extended using the Pin extension for open tray impresson components.
- The tightening torque of the impression components is 5–10 Ncm – light finger force.